
































Continuity 



















Range 



20/200/2000Ω (Autoranging) 























Open circuit voltage (DC) 



5V±20% (*1) 























Short circuit current 



Greater than 200mA 























Accuracy 



±0.1Ω (0  0.19Ω) ±2%rdg+8dgt (0.2  2000Ω) 















Insulation Resistance 





















Range 



20/200MΩ (Autoranging) 



20/200/2000MΩ (Autoranging) 





























Open circuit voltage (DC) 



250V+25% 0% 



500V+25% 0% 



1000V+20% 0% 































Rated current 



1mA or greater @ 250kΩ 



1mA or greater @ 500kΩ 



1mA or greater @ 1MΩ 





























Accuracy 



±2%rdg+6dgt (0  19.99MΩ) ±5%rdg+6dgt (20  200MΩ) 



±2%rdg+6dgt (0  199.9MΩ) ±5%rdg+6dgt (200  2000MΩ) 

















Loop Impedance 























Function 



LPE 



LPE (ATT) 



LN / LL 





























Rated Voltage 



100  260V 50/60Hz 



LN: 100  300V LL: 300  500V 50/60Hz 





























Nominal Test Current at 0Ω External Loop: Magnitude / Duration (*2) 



20Ω: 6A/20ms 200Ω: 2A/20ms 2000Ω: 15mA/500ms 



LN: 6A/60ms NPE: 10mA/approx. 5s 



20Ω: 6A/20ms 































Range 



20/200/2000Ω (Autoranging) 



20/200/2000Ω (Autoranging) (LN < 20Ω) 



20Ω 































Accuracy 



±3%rdg+4dgt (*3) ±3%rdg+8dgt (*4) 



±3%rdg+6dgt (*3) ±3%rdg+8dgt (*4) 



±3%rdg+4dgt (*3) ±3%rdg+8dgt (*4) 



















PSC (LN / LL) / PFC (LPE) 























Function 



PSC 



PFC 



PFC (ATT) 































Rated Voltage 



100  500V 50/60Hz 



100  260V 50/60Hz 



 































Nominal Test Current at 0Ω External Loop: Magnitude / Duration (*5) 



6A/20ms 



6A/20ms 2A/20ms 15mA/500ms 



LN: 6A/60ms NPE: 10mA/approx. 5s 



























Range 



2000A/20kA AutoRanging 























Accuracy 



PSC / PFC accuracy is derived from measured loop impedance specification and measured voltage specification 















RCD 



























Function 



X1/2 



X1 



X5 



Ramp 



Auto 





































Trip current setting 



10/30/100/300/ 500/1000mA 



10/30/ 100mA 



10/30/ 100/300/ 500mA 



Depending on the accuracy at each function. Measurement sequence: X1/2 0°→X1/2 180° →X1 0°→X1 180° →X5 0°→X5 180° Measurements with x5 are not carried out for RCDs with nominal current of 100mA or more. 




































Trip current Duration 



2000ms 



G: 550ms S: 1000ms 



410ms 



Goes up by 10% from 20% to 110% G:300ms/ S:500ms X10 times 

































Rated Voltage 



230V+10%15% 50/60Hz 































Accuracy 



AC Type: 8%  2% A Type: 10%  0% 



AC Type: +2%  +8% A Type: 0%  +10% 



AC Type: ±4% A Type: ±10% 































Trip time: ±1%rdg+3dgt 


















RCD (Uc) 



















Function 



UC 























Rated Voltage 



230V+10%  15% 50/60Hz 























Range 



100.0V 























Test current 



≤ 1/2IΔn (max150mA) 























Accuracy 



+5%  +15%rdg, ±8dgt 















Earth 



















Range 



20/200/2000Ω AutoRanging 























Accuracy 



20Ω: ±3%rdg+0.1Ω 200/2000Ω: ±3%rdg+3dgt (Auxiliary earth resistance 100±5%) 















Phase Rotation 



















Rated voltage 



50500V 50/60Hz 























Remarks 



Correct phase sequence: are displayed "1.2.3" and mark Reversed phase sequence: are displayed "3.2.1" and mark 















Volts 





















Function 



Volts 



Frequency 

























Rated voltage 



25  500V, 45  65Hz 

























Measuring range 



25  500V 



45  65Hz 



























Accuracy 



±2%rdg+4dgt 



±0.5%rdg+2dgt 

















General 



















Applicable standards 



IEC 610101 CAT III 300V (500V L to L) Pollution degree 2, IEC 61010031, IEC 615571, 2, 3, 4, 5, 6, 7, 10, IEC 60529 (IP40), IEC 61326 (EMC) 























Power source 



LR6 X 8 























Dimensions 



136(L) X 235(W) X 114(D)mm 























Weight 



1350g (including batteries.) 























Accessories 



7188A (Distribution board fuse test lead set) 7196A (Test leads with remote control switch) 7218A (Power cord [EU]) (*6) 7228A (Earth resistance test leads) 8032 (Auxiliary earth spikes) 8212USB (USB adaptor with KEW Report (Software)) 9142 (Carrying case) 9121 (Shoulder strap) Cord case, Buckle, Battery, Instruction manual 























Optional 



8212RS232C (RS232C adaptor with KEW Report (Software)) 









